Linshang new UV irradiance meter LS130 is specially designed to test the intensity and energy of high-pressure mercury lamps, halogen lamps and other light sources. It can also display the intensity curve. The power real-time value, power maximum value, energy value and measurement duration can be displayed simultaneously during the test. The length of UV curing time can be known by the measuring duration.
So how to use UV irradiance meter LS130 ?
First of all, customers can choose automatic or manual measurement according to their needs. If the production line is very long, it will take a long time to reach the equipment from the UV source. It is recommended to enable the automatic measurement mode. If it is used routinely, it is recommended to enable the manual measurement mode. After confirming the mode, it will enter the historical data query interface.
When measuring, press the “POWER” button to start a new test. After selecting “YES”, start a new measurement. Align the detector of the meter with the UV light source and start the test. The power real-time value, power maximum value, energy value and measurement duration will be displayed on the screen. The test is completed (in automatic mode, when the light source intensity is less than the trigger threshold, the instrument will automatically stop the measurement. In manual mode, you need to manually press the “POWER” button to stop the measurement).
After the test is completed, press the “ENTER” button to view the measured power curve.
The UV irradiance meter LS130 can also be connected to a computer via a USB interface to read detailed test reports and print test reports.