Gloss Measurement of Waterborne Epoxy Coatings
1. Factors Influencing Waterborne Epoxy Gloss
Water-based epoxy coatings have the advantages of high hardness, strong adhesion and good chemical resistance. They are widely used in anticorrosive coatings and industrial floor coatings. However, different waterborne epoxy formulations and construction processes can cause differences in gloss of waterborne epoxy paint films.
(1) Production formula of water-based epoxy floor
Waterborne epoxy coatings are made by mixing standard liquid epoxy resin, water, defoamer, wetting agent and waterborne epoxy curing agent. Different formulations will cause a certain difference in gloss of waterborne epoxy paint films.
(2) Coating thickness and uniformity
The uniformity of the coating film mixing and the application coating thickness will cause differences in gloss of the water-based epoxy paint film.
(3) Flatness of substrate
If the floor is not polished uniformly and there are holes, it will also affect the gloss of the water-based epoxy paint film.
2. Method for measuring gloss of water-based epoxy coatings
(1) Gloss detection of waterborne epoxy coatings
Through the Linshang portable gloss meter, you can accurately and quickly determine the surface gloss of water-based epoxy paint films. Gloss is a visual property of an object. The more light that can be reflected from the object surface, the more pronounced the gloss and the higher the gloss. Judging the object surface gloss with the naked eye depends entirely on experience, but to truly evaluate the size and uniformity of the gloss, you need to use a professional portable gloss meter to measure it.
(2) How to choose the gloss measurement angle of waterborne epoxy coating?
The gloss measured by the portable gloss meters is related to the angle of incidence of light. So when testing the same material, the test data of the portable gloss meter angle is different. 60 degrees is the most versatile portable gloss meter angle. Any gloss material can be measured using the standard measurement angle of 60 degrees. High gloss, medium gloss and low gloss are also divided according to the test data of the 60 degree portable gloss meter.
LS191 Portable Gloss Meter
Linshang gloss measuring instruments are all independently developed and produced. The universal 60 ° measurement angle is suitable for the detection of the surface gloss of low and medium gloss materials. It is also more suitable for gloss testing of waterborne epoxy coatings.
For the measurement and verification of glossiness measuring instruments, the country has clear regulations. The standard is "JJG696-2015 Specular Gloss Meters and Gloss Plates". All indicators of the Linshang gloss measuring instrument have met the requirementsof the first-class working machine in the national metrological verification regulations.
The portable gloss meter produced by Linshang can be said to be a better brand among Chinese portable gloss meters. It's perfect to use it to check the specific gloss of waterborne epoxy coatings!