UVLightMeter APP is the supporting mobile phone application of Linshang Technology LS126C UV light meter. Through the Bluetooth communication and the UV meter APP provided by the instrument, the user can monitor the intensity of the germicidal lamp tube in real time, avoiding the radiation damage caused by ultraviolet rays in the field test.
1. Mobile UV meter APP installation
UVLightMeter APP supports Android OS 5.0 and above, you can choose one of the following methods to install.
1) Use the mobile browser or WeChat to scan the QR code below, and follow the prompts to download and install the UVLightMeter.apk software.
2) Copy the UVLightMeter.apk software from the accessory USB drive to your phone and follow the prompts to install it.
After the installation is complete, the “UVLightMeter” icon will appear .
2. Mobile UV meter APP operation
1) The first time you use the app, the operation is as follows:
a. Short press the UV light meter power button to turn the instrument on.
b. Click the “UVLightMeter” icon on the mobile phone desktop to run the app and the welcome screen appears.
c. After the interface stays for 3 seconds, enter the “Bluetooth” interface, as shown in “Left” below. Click "Start Search" and the app starts searching for devices. When the device is searched, the “Other Available Devices” column will list all the searched device names, as shown in the following figure (right), click on the device whose initial letter is LS, the last 9 digits are the same as the instrument SN number, APP This instrument will be bound and jumped to the measurement interface.
(Note: The SN number of each instrument is viewed on the label on the back of the instrument)
Measurement interface, UV meter APP automatically connects to the instrument. After successful connection, the instrument model, SN number, real-time power value (P), statistical maximum value (PMAX), minimum value (PMIN), average value (PAVG) are displayed, as shown in the figure below.
Click “Re-count”, the instrument clears the current data and starts a new measurement.
2) The device is already bound, and the operation is as follows:
After the UV meter APP connection is successful, the instrument will be automatically bound. Once again, the APP will directly enter the measurement interface to automatically connect to the last connected instrument. After successful connection, the instrument model, SN number, real-time power value (P), and statistical maximum value will be displayed. (PMAX), minimum (PMIN), average (PAVG).
The APP can only bind one instrument at a time. If you need to bind another instrument, you must enter the “Bluetooth” interface, click “Start Search”, and select the new instrument in the “Other Available Devices” column of the search results.
More product information please refer to "LS126C UV light meter"