The hospital is a place with a lot of germs. If it is not sterilized in time, it may cause bacterial infection. Therefore, ultraviolet sterilization is important at this time. The ultraviolet radiation intensity directly affects the ultraviolet sterilization effect. So how do you detect the radiation intensity of UV germicidal lamps? Generally, we use a UV lamp intensity meter for testing. The detection methods are also different for different types of UV lamp intensity meter. Today we take the Linshang LS126C UV lamp intensity meter as an example to find out how to detect the radiation intensity of ultraviolet germicidal lamps.
1)Hospital germicidal lamps generally require testing one meter below the lamp. So the instrument is equipped with a one-meter hook and a piece of iron underneath the hook.The probe is magnetically self-adhesive on the iron plate. Then press the power button to set the parameters you need and confirm, the instrument will enter the historical data query interface.
2)Short press the power button to start the test, then hang the test hook under the lamp. The person leaves and turns on the germicidal lamp for testing. The instrument will automatically shut down after 10 minutes.
3)Short press the power button to start the machine and the instrument enters the historical data query interface. The interface will display No.1 (last test data), power graph during measurement. Real-time value, maximum, minimum value, average value, measurement duration, date and time of measurement of last time will be displayed.
4)When the instrument enters the historical data query interface, you can scroll up and down to view the historical data. The instrument can record a total of 25 sets of data.
2.Mobile APP operation method:
1)Install and download the mobile app.
2)Short press the power button to turn on the phone and check the SN number when the phone is turned on.
3)Open the mobile app and click on the upper right corner to start searching. The search result will display the SN number that is consistent with the instrument. Click on the searched device and the instrument will jump to the measurement interface. The connection is successful in the upper right corner.
4) Place the instrument on the measuring iron plate with the hook and press the measurement button. Leave the measurement site and turn on the germicidal lamp to start the test. We can monitor the data in real time on the mobile app. The maximum, minimum, average and real-time values are displayed on the APP. The Bluetooth real-time transmission distance is 50 meters in the open area and the APP has a re-statistic function.
5) The instrument is equipped with computer software, users can print out test data and test reports according to their own needs.