Difference between LS116 Light Transmittance Meter and LS117 Optical Density Meter
Customers who often buy Linshang instruments can't tell the difference between the LS116 light transmittance meter and the LS117 optical density meter. They feel that these two products are similar, but this is not the case. In this article, we will mainly introduce the difference between the LS116 light transmittance meter and the LS117 optical density meter.
|Model||LS116 light transmittance meter||LS117 optical density meter|
|Testing material||glass, solar film,transparent and uniform regular transmission materials||
Opalescent, fog and frosted materials such as aluminized sheet, X-ray film, film, lens ink
|Material size||Longer than 10 mm in diameter and thinner than 100 mm in thickness||Longer than 2 mm in diameter|
|Transmitted light intensity/incident light intensity||OD=log10(incident light/transmitted light) or OD=log10(1/light transmittance)|
|Optical path||Parallel optical path design||Diffuse transmission path with integrating sphere effect|
Compared with the LS116 light transmittance meter,the LS117 optical density metercan measure not only the transmittance of a diffuse transmission material but also the transmittance and optical density of a low transmittance material.And it has more powerful function.Therefore,the price of the two instruments is also different.The LS116 light transmittance meter cost $292 and the LS117 optical density meter cost $550.The above is the main difference between the LS116 light transmittance meter and the LS117 optical density meter.
- Difference between LS116 and LS117 Light Transmittance Meter
- What’s the Difference Between Point Light and Parallel Light Transmittance Meter
- Measurement of Optical Density
- Spectacle lens anti-blue light detection---blue-violet light transmittance meter
- Difference of LS116 Transmission & LS117 OD Meter
- Linshang Insulated Glass Unit Measuring Tools