Factors Affecting the Gloss of Steel Plates and Measurement Methods | Gloss Meter
The gloss of the steel plate refers to the specular reflection ability of the surface of the steel plate to light. In order to accurately evaluate the gloss of the steel plate, it is often measured by a gloss meter.
1. What is the gloss of the steel plate?
Rigid board gloss is a set of physical quantities that evaluate the ability of the steel plate to reflect light under geometrically prescribed conditions. With direction-selective reflective properties, we usually refer to gloss as "mirror gloss." In simple terms, the gloss of a steel plate refers to the reflective ability of its surface.
Metals are metal crystals combined by metal bonds. Metal atoms are arranged in the most densely packed state. There are free electrons inside the metal. Its range of motion is the entire metal. When white light hits the metal surface, free electrons can absorb all Light with a wavelength is then reflected again, so most metals (except gold is yellow, copper is red, cesium is light yellow, bismuth is light red, and lead is light blue) have a silvery white luster.
2. What are the factors that affect the gloss of the steel plate?
The gloss of different steel plates is mainly related to the characteristics of the reflected light on the surface. For example, if the metal surface is kept shiny, other oxides may be formed without metallic luster; and diffuse reflection is formed with less metallic luster. Wire cutting, laser, and high cutting temperature may cause oxides and other compounds on the surface of the steel plate to cause little or no metal luster.
3. How to improve the gloss of the steel plate?
Generally speaking, the smaller the surface roughness of the steel sheet, the higher the surface glossiness, and vice versa. Therefore, the gloss of the steel plate can be improved by polishing.
Polishing refers to a machining method that reduces the surface roughness of an object by mechanical, chemical or electrochemical methods to obtain a bright and flat surface. Polishing cannot improve the dimensional accuracy or geometric accuracy of the workpiece, but is used for the purpose of obtaining a smooth surface or specular gloss, and sometimes to eliminate gloss (matting).
In addition, steel plate coating is also a commonly used method to improve gloss. By coating a thin layer on some metal surfaces, it can have a good protection effect, prevent metal oxidation (such as rust), improve wear resistance, conductivity, light reflectance, corrosion resistance (copper sulfate, etc.) and improve Aesthetic effect.
4. Measuring gloss of steel plate with gloss meter
Gloss is a relatively common reference indicator in industry. For example, metal processing, ceramic products, paints and coatings, textile dyeing, etc. will have gloss requirements. In the actual production process, in order to accurately evaluate the gloss of the steel sheet and ensure the appearance quality of the product, it can be measured by a gloss meter. Gloss meter is a special instrument for measuring the surface gloss of objects.
At present, the domestic gloss meter meets the requirements of the JJG working gloss meter before it can be used normally. Like the LS192 gloss meter, a 60-degree angle light source is projected, and it is placed on the surface of the steel plate during measurement. According to the conditions of the light source and the receiver, the light flux of the steel plate in the specular reflection direction and the reflection of the glass standard in the specular reflection direction The ratio of light to luminous flux is converted into data by electronic calculation and displayed on the LCD screen of the instrument. The display interface of the LS192 gloss meter has a statistical function, which can display the maximum, minimum, average and standard deviation of the test. This feature can be used to analyze the uniformity of the surface gloss of the steel plate.
- Glossiness Measurement and Gloss Meter Standard
- Working Principle and Measuring Range of Magnetic Thickness Gauge
- How to Choose a Film Thickness Gauge?
- Factors Affecting the Eddy Current Thickness Gauge Accuracy
- Mist Material Optical Transmission Meter
- Why should We Use Exposure Machine UV Integrator?