Feedbacks from Customer about LS181 Tint Meter for Sale
Some time ago, one customer responded with two questions when using the LS181 tint meter for sale.
1.The data of the LS181 is inaccurate and unstable. There are deviations about 5% when the material is placed on the left and right.
2.The data will also jump when using under strong light.
The LS181 tint meter for sale is an upgraded version of the old LS101. It upgrades the light source and power supply mode. The parallel light design avoids the refraction of the scattered light source. It is more accurate when testing thick materials and has a wider card slot. 5v power supply is also recognized as the most stable power supply method in the world and the tint meter can be used in most of the charge pal on the market.
Open the customer video and watch, the customer is moving back and forth with the film in the middle of the LS181 card slot. In fact, this situation is normal, the test card slot of the LS181 is relatively wide, the customer's hand is shaking and the solar film is softer. So the measured data must be jumping back and forth.
When tested under strong light, the data will also jump. That is because the optical instruments are sensitive. In the strong light or the ash inside the light hole will affect the data accuracy, so the measurement environment is very important.
- Anti-blue Glass Lens Tester, Myopia People Should Pay Attention
- New Method of IR Ink Transmission Meter
- How to Maintain the Anti-blue Transmission Meter
- Compare Test Data of Linshang LS183 and LS108H Optical Transmission Meter
- LS183 Thermal Insulation Coating Tester
- Why Use LS108A and LS108D Mobile Phone Lens IR Ink Hole Detection